Hi, has anyone else experienced the 5FR Powerpicc splitting approx 5 cm below the connector whilst flushing or infusing? We just had a bad extravasation of oxaliplatin - the line had split below the insertion site, in the vein. It's not possible that it could have been damaged as it was internal. This is the fourth event since I started using them about a year ago (but fortunately the others weren't chemo). It seems like there is a problem!