This product is already available and it's called MaxPlus clear from Maximus.
lme PICC Team Clinical Leader
Is anyone having trouble with the Microclave?
That comment could be quite misleading to some folks not as familiar with all of the discussion about needleless connectors. The MaxPlus clear and the Microclave are distinctly different connectors with different technical specs.
Please don't add to the confusion.
Mari Cordes, BS RN
Mari Cordes, BS RNIII VA-BC
Vascular Access Department
University of Vermont Medical Center
You're right. They are quite different. The microvlave is negative displacement and can't be cleaned very well before access because of the large (compared to bactera) gap on the surface where the syringe touches. The maxplus is positive displacement and can be cleaned very well before access. There's a study out there that says the clave/microclave doesn't pass this bacteria through it, but the 1st flush in the study isn't recorded, and this is where the bacteria gets flushed through. I don't understand why this sample wasn't measured.